|Brief description: || |
In order to determine this with the Lauda TD 3 tensiometer, a testing cylinder is filled with 20 ml each of distilled water and the transformer oil to be tested. Due to the huge differences in polarity, two phases are rapidly formed. Water, which has the higher density, forms the lower phase and the insulating oil floating above. Fresh transformer oil is non-polar. Polar components have formed in the used oil, which accumulate in the region of the barrier layer of water (polar) and oil (non-polar).
The interfacial tension of the water/oil phases is influenced by the interaction of the ageing products with the water. The Lecomte Du Noüy ring, secured by a thread to a scale, is lowered into the water-oil mixture. By lowering the cylinder, the ring is first drawn through the water and then through the interfacial into the oil. The force occurring during phase change at the interfacial is measured.